A theodolite is used for measuring vertical and horizontal angles. The main parts are the horizontal and the vertical circles, which are scaled in gon. It is regularly placed on a tripod and features several spirit levels for adjusting the theodolite in the field.
A theodolite featuring a device for optical distance measurement is called tachymeter. More modern tachymeters equipped with electrical distance measurement devices are called total stations.
Measuring polygonal chains
The choice of the points (vertices) of the polygonal chain depend on the aim of the mensuration. In most cases, one will follow the course of roads and trails. The single segments should be located in a way that other objects can be connected easily. The line segments should be relatively long, to keep the error propagation low - not less than 50 meters and, under normal circumstances, not more than 200 to 300 meters .
To asses the absolute position of a polygon chain, it has to be connected to a trigonometric vertex close to it. Generally, close vertices should be part of the polygonal chain.
In the field, the procedure goes as follows: The individual vertices are to be marked with perpendicular ranging rods. The theodolite is centered exactly over the respective vertex. Both tasks are to be fulfilled with full attention to minimize errors in the angle measurement.
From each vertex, the distance to the preceding and to the following vertex is measured. The distance mensuration is done twice, the second measuring can be seen as safety mensuration. The mean of both mensurations is taken for the best approximation of the real distance. The distance is mostly obtained with electronic distance measurement tools. The angle of refraction between two distances can be derived from the difference between the two readings of the horizontal circle, obtained by the bearing to the preceding and to the following vertex.
The angle of inclination is derived from the vertical circle reading. All measured parameters are recorded on a survey sheet. Additionaly, all vertices are plotted in a sketch map.
Polar coordinate method
This method can be applied in clearly arranged terrain. Measurement is conducted from one fixed point. The distance and the angle of refraction from the point are measured. Thus, it is basically a measurement of polar coordinates. The distance is taken tachymetric or electronic. To draw the obtained vertices to a map, the coordinates of at least two points have to be available.
Today, this is the standard method for extensive measurements, as angle and distance measuring can be conducted automatically in most cases .